hi guys, my form 2 optical window was cloudy inside, so i decided to clean it. the problem is i unscrew the tank carrier screws also… (i know doomed)
trying to re calibrate now. is there any official PDF for that ?
right now im pretty close with my 10mm high test cubes.
the front screws all the way screwed in, and i playing with the back side screws adjustments, and get this result.
but sadly when i printed the official OpticsTest.form i have got horrible result.
i know, i should not use engineering resin for calibration because it is not optimized for that, but right now i only have that to work with. (trying to get standard resin ASAP)
my question is, could be the failed print because the resin i use, calibration, or something else ?